1.800.431.4310 nascohealthcare.com Email: [email protected]
COPYRIGHT © NASCO 2010 PRINTED IN U.S.A. NP 072-11/RV 02-12
About the Simulator
The Nasco Life/form® Pediatric Lumbar Puncture Simulator represents a 12-month-old child in the left lateral decubitus position. The embedded iliac crest offers exceptional realism, while the removable spine, spinal canal, and skin pad make training simple and hassle-free. Students will appreciate the opportunity to practice this delicate, but commonly-performed, procedure on a manikin that is both appealing and anatomically accurate.
CAUTION: PRODUCT CONTAINS DRY NATURAL RUBBER!
List of Components
- Body on Base Board
- Tissue Block
- Skin Pad
- Tubing Assembly
- IV Fluid Supply Bag
- Baby Powder
Setting Up the Trainer
- Feed the straight section of the tubing assembly through either end of the spine, seating the latex portion within the spine. (See figure 1.)
- Snap the spine into the tissue block. (See figure 2.)
- Feed the left-hand tubing section (the one attached to the elbow fitting) through the red fitting at the top of the body cavity and out the top of the body. (See figure 3.)
- Insert the tissue block assembly into the cavity while pulling the tubing through.
- Snap the skin pad over the cavity. A light dusting of baby powder will give the skin a more realistic feel. (See figure 4.)
- Connect the left-hand tubing section to the IV bag. (See figure 5.)
- Close the IV bag clamp and fill the bag with about 200 cc of water. Hang the bag about 16" from the table surface. (Hanging height and water volume will determine the fluid pressure within the spinal canal and can be adjusted as desired.)
- Connect the luer barb and stopcock to the other end of the tubing. Set up a towel or container sufficient to absorb or collect a small amount of water from the stopcock. (See figure 6.)
- Open the clamp and the stopcock. Allow water to flow until the air is displaced. Close the stopcock only – leave the clamp open.
Performing the Procedure
- The lumbar puncture may be performed in the L3-L4, L4-L5, or L5-S1 space.
- Locate the correct puncture site by palpating the iliac crest and spine.
- After each procedure, pass a small amount of water through the stopcock to eliminate any air pockets that may have formed when fluid was withdrawn. Refill the IV bag as needed to maintain correct pressure.
Care and Maintenance
- The skin pad and the latex spinal tubing should be replaced when they are no longer usable. Varying the puncture site, rotating the tubing, and using the smallest needle consistent with correct protocol will extend the life of the parts. Dull or burred needles will cause leakage and premature wear.
- Always drain all fluids before storing the unit.
- The tissue block and skin pad may be washed with soap and warm water to remove dirt or oils. Inks, markers, and some antiseptic solutions may permanently stain the plastic.
- The body may be cleaned with mild soap or detergent and a damp cloth, or with Nasco Cleaner. Do not submerge the body in water.
Available SuppliesLF01000(A)U – Pediatric Lumbar Puncture Replacement Kit
LF09919U – Nasco Cleaner